At the end of this chapter, the learner will be able to:
Define terms describing the use of ultraviolet C including electromagnetic spectrum, ultraviolet radiation, minimal erythemal dose (MED), and the cosine law.
Explain how the application of ultraviolet C to a chronic wound generates a bactericidal effect.
Develop safe and appropriate application parameters for ultraviolet C in wound healing.
Select specific patient indications appropriate for the use of ultraviolet C in wound management.
Identify precautions and contraindications for the use of ultraviolet C.
Select safe and appropriate parameters for ultraviolet C application.
Develop and implement a care plan involving the use of ultraviolet C in the treatment of a chronic wound.
Although ultraviolet A (UVA) and ultraviolet B (UVB) have been used in health care for some time to treat dermatological conditions such as psoriasis and also have evidence to support the use of ultraviolet radiation to promote wound healing, ultraviolet C is the type of ultraviolet light that is most commonly utilized today in the treatment of chronic wounds1, 2. Specifically, ultraviolet C (UVC) is beneficial for the treatment of chronic wounds due to its bactericidal effect.3 As the number of drug-resistant organisms continues to increase, wound clinicians must consider treatment options that not only effectively destroy bacteria without damaging fibroblasts and other cells necessary for wound healing, but also avoid the development of resistance.4 Ultraviolet light C accomplishes both of these treatment goals and is a cost-effective, portable, and safe treatment modality.5, 6
Understanding the types of ultraviolet light is important in order to choose a wavelength that will deliver the desired outcome (TABLE 19-1, FIGURE 19-1). This chapter focuses on ultraviolet C, which has FDA approval for use on open wounds due its bactericidal effects.7, 8 TABLE 19-2 provides terminology and definitions utilized in the discussion of ultraviolet light. Handheld ultraviolet C devices consist of lamps that deliver ultraviolet light at a specific wavelength (254 nm), which falls within the optimal wavelength range for bacterial reduction.9, 10 These devices contain filters specific to UVC that reduce the risk of skin cancer and skin burns often associated with UVA and UVB.11 Although studies have shown an increase in risk of skin cancer associated with the use of ultraviolet B, there has been no link reported between skin cancer and UVC.11, 12
Table 19-1Comparison of Ultraviolet A, B, and C18 |Favorite Table|Download (.pdf) Table 19-1 Comparison of Ultraviolet A, B, and C18
|Ultraviolet Light Characteristics ||Ultraviolet A ||Ultraviolet B ||Ultraviolet C |
|Wavelength ||320-400 nm ||290-320 nm ||100-290 nm |
|Effects || |
Log In to View More
If you don't have a subscription, please view our individual subscription options below to find out how you can gain access to this content.
Want remote access to your institution's subscription?
Sign in to your MyAccess profile while you are actively authenticated on this site via your institution (you will be able to verify this by looking at the top right corner of the screen - if you see your institution's name, you are authenticated). Once logged in to your MyAccess profile, you will be able to access your institution's subscription for 90 days from any location. You must be logged in while authenticated at least once every 90 days to maintain this remote access.
If your institution subscribes to this resource, and you don't have a MyAccess profile, please contact your library's reference desk for information on how to gain access to this resource from off-campus.
AccessPhysiotherapy Full Site: One-Year Subscription
Connect to the full suite of AccessPhysiotherapy content and resources including interactive NPTE review, more than 500 videos, Anatomy & Physiology Revealed, 20+ leading textbooks, and more.
Pay Per View: Timed Access to all of AccessPhysiotherapy
24 Hour Subscription $34.95
48 Hour Subscription $54.95
Pop-up div Successfully Displayed
This div only appears when the trigger link is hovered over.
Otherwise it is hidden from view.